- Leading experts slam current diabetes care in UK care homes – a Call to Action needed
- Roche announces urgent recall on certain Accu-Chek Aviva and Performa test strips
- Ramadan 2018 and fasting
- DVLA changes regulations
- Freestyle Libre available on the NHS!
- Medtronic recall specific lots of infusion sets with all models of Medtronic insulin pumps
- All Accu-Chek® Insight insulin pumps – risk of alarm failure
- WHO – long-acting analogues not an essential medicine
- Beef insulin discontinued
- Soft Drinks Warning!
- Driven to Despair
- 2017 – Getting back on track after Christmas
- Childhood Obesity Strategy failing our children
- Diabetes – What schools need to know
- DVLA – Good News!
- The National Diabetes Foot Care Audit (NDFA)
- Diabetes care criticised
- Motorists banned in error after faulty DVLA visual field test
- Too many children and young people with diabetes not getting the care they need
- National NHS Diabetes Prevention Programme
- InDependent Diabetes Trust: Young diabetics ‘get worse care’
- IDDT’s Position Statement – ‘pre-diabetes’
- MBE Is Bolt From The Blue For Charity Chair
- New insulin on the market – Insulin Degludec (Tresiba)
- Diabetes community urges more support for older people
- Warning to people with diabetes about dangerous herbal medicine
- In 1 week, 60 hospital patients with diabetes develop preventable complications – National Audit
- Passport for Diabetes in Care Settings
- Launch of Passport for Diabetes in Care Settings
- Type 2 Diabetes – Management & Medication
- Cost of insulin analogues
- Actos And The Risk Of Bladder Cancer – New Safety Warnings
- Actraphane – are patients really at the centre of care
- Marketing of insulin – a missed opportunity
- IDDT Launch Patient Hospital Passports
- IDDT Launch Public Awareness Campaign
- IDDT’s Position Statement on DTCA
- In Sickness and in Health
- IDDT Triumphs in Australia
Roche announces urgent recall on certain Accu-Chek Aviva and Performa test strips
May 11th 2018
Roche Diabetes Care have asked for a recall on certain batches of its Accu-Chek Aviva and Performa test strips. The affected batches could result in readings that are either too high or tool low which in turn, can lead to treatment errors.
The company report that some strips in these batches may be giving error messages when inserted into the blood glucose meter, whilst other test strips may not give error messages but may still give the wrong readings.
If you are using Accu-Chek Aviva or Accu-Chek Performa test strips, please check whether you have any test strips in the batches affected.
Affected lot numbers for Accu-Chek Aviva test strips
The lot numbers can be found on the top of the boxes of test strips and on the side of the test strip tubs.
50 test strip pots:
10 test strip pots:
Affected lot numbers for Accu-Chek Performa test strips
10 test strip pots:
Test strips in the Accu-Chek Performa Nano blood glucose kit:
What to do if your test strips are affected
If your test strips are affected, it is important that you stop using them. If you have some unaffected test strips, use those instead.
Please take any affected lots back to the pharmacy they came from. The pharmacy may be able to give a replacement pack or otherwise advise you on what to do.
If you have questions about the recall, please call Accu-Chek on 0800 701 000.